Fast point-of-use sterilization makes turnaround a priority, increasing efficiency and virtually eliminates instrument bottlenecks. Our state-of-the-art steam technology is gentle on both solid and hollow instruments. Air Removal Using PPPD technology and our patented steam injection process — air is automatically forced out of the sterilization chamber. Touch A large 3.
|Published (Last):||21 July 2017|
|PDF File Size:||9.52 Mb|
|ePub File Size:||11.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fast point-of-use sterilization makes turnaround a priority, increasing efficiency and virtually eliminates instrument bottlenecks. Our state-of-the-art steam technology is gentle on both solid and hollow instruments.
Air Removal Using PPPD technology and our patented steam injection process — air is automatically forced out of the sterilization chamber. Touch A large 3. Fully adjustable contrast levels let you optimize the screen to the lighting in your steri-centre. Connect Through your networked portal, you can vie not only your STATIMs current operations in real time, you can also access cycle history, printing and validation information, and also sterilization records.
Protect Patient safety relies on attention to detail. This encourages extra attention to process and creates a safer environment for both staff and patients. Collect An undocumented sterilization cycle might as well not have been performed. With each cycle, information must be stored to protect your office and your patient. The integrated DataLogger stores all cycle information for the lifetime of the unit, which is accessible through multiple channels such as: a USB port for manual retrieval; an Ethernet connection for web based archiving or desktop printing; or a RS port for an external printer option.
Cassette autoclave Cycle times: 9 minutes unwrapped — If you do not find the problem below or the action you take does not solve the problem, please contact our support team on 10 20 Unit does not power ON Action Check that the unit is plugged into a properly grounded outlet and that the power cord is firmly seated at the rear of the machine.
Try another circuit. Check the condition of the line circuit breaker or fuse. There is water under the machine Action Check that water was not spilled when refilling the reservoir. Make sure the plug in the exhaust tube is secured. Remove and reinsert the cassette. Attempt another cycle. Be careful. The cassette is leaking.
If water drips from the underside of the unit during operation, check the cassette seal for misalignment or damage and replace the seal if required. If it still leaks attempt another cycle using a different cassette if possible. Instruments do not dry Action Best drying occurs when the cycle continues to completion. Allow the cycle to finish. Make sure the instruments are loaded correctly in the cassette. Refer to section 4. Check the unit levelling. Refer to section 8. Examine the exhaust tube tube to the waste bottle for kinks.
If kinked, straighten the tube. Depress the collar on the fitting and with the other hand pull firmly on the tube. Once the tube is free of the fitting, cut the damaged section of tubing away using a sharp instrument. Be sure that you leave enough tube to reach the unit when you re-attach the tube to the exhaust fitting. Make sure the compressor is working. To check, remove the exhaust tube from the waste bottle. If there is not a strong steady flow of bubbles, the compressor is not functioning properly.
Call our service team on Remove the cassette. The metal parts will be hot and the cassette will contain hot steam. Check the exhaust tube for kinks or obstructions. Refer to section 3. Try running another cycle. If the problem persists, record the cycle fault message number and contact our service team on Excessive steam from the front of the machine Action Remove and reinsert the cassette.
Remove and check the cassette seal for misalignment or damage. Replace the seal if required. Be careful as the metal parts will be hot and the cassette will contain hot steam. If the leak persists, turn the unit OFF, remove and unload the cassette and contact our service team on Machine will not start and touchscreen displays H2O X icon Action Press on the icon to confirm whether it is a water level problem or a water quality problem.
If it is a water quality problem, you have likely used water which is not steam-process distilled or is improperly distilled. If the level of the water in the reservoir is low, refill the reservoir. The impurities and additives in other water sources will cause an error reading on the LCD. If you have the water conductivity meter, check the quality of the water before refilling the reservoir.
Make sure that the printer is powered ON. Time and date are incorrect Action The time and date have not been set. See section 3. Power off the unit and power it on again. It can take up to 6 minutes before the main menu screen appears.
Statim 5000 Repair Parts
To make a correction, select the PIN User you want to change and repeat the process described above. To activate Process Enforced usage, follow these steps: 1. Scroll to Process Enforced 3. Press to the Setup menu. To change themes follow these steps: 1. Scroll to Themes and select.
Scican Statim 5000 Manuals
Tests conducted by leading universities in North America and Europe have proven the Statim effectiveness for dental and medical sterilization. Additionally, it is easy to use and maintain. Simply select one of the three sterilization programs and with a single touch of the keypad, begin the cycle. An advanced microprocessor controls all functions and maintains optimal sterilization conditions, from start to finish.
Free Statim 2000 Operational Manual